By a News Reporter-Staff News Editor at Health & Medicine Week -- Researchers detail new data in Eye Diseases and Conditions. According to news reporting originating in La Jolla, California, by NewsRx journalists, research stated, "To evaluate the diagnostic accuracy of spectral domain optical coherence tomography (SD-OCT) for detection of preperimetric glaucoma and compare it with the performance of confocal scanning laser ophthalmoscopy (CSLO). Cohort study. A cohort of 134 eyes of 88 glaucoma suspects based on the appearance of the optic disc."
The news reporters obtained a quote from the research from the University of California, "Patients were recruited from the Diagnostic Innovations in Glaucoma Study (DIGS). All eyes underwent retinal nerve fiber layer (RNFL) imaging with Spectralis SD-OCT (Heidelberg Engineering, Carlsbad, CA) and topographic imaging with Heidelberg Retinal Tomograph III (HRT-III) (Heidelberg Engineering) CSLO within 6 months of each other. All patients had normal visual fields at the time of imaging and were classified on the basis of history of documented stereophotographic evidence of progressive glaucomatous change in the appearance of the optic nerve occurring before the imaging sessions. Areas under the receiver operating characteristic curves (AUCs) were calculated to summarize diagnostic accuracies of the SD-OCT and CSLO. Likelihood ratios (LRs) were reported using the diagnostic categorization provided by each instrument after comparison to its normative database. Forty-eight eyes of 42 patients had evidence of progressive glaucomatous change and were included in the preperimetric glaucoma group. Eighty-six eyes of 46 patients without any evidence of progressive glaucomatous change followed untreated for an average of 14.0 ± 3.6 years were included in the control group. The parameter with the largest AUC obtained with the SD-OCT was the temporal superior RNFL thickness (0.88 ± 0.03), followed by global RNFL thickness (0.86 ± 0.03) and temporal inferior RNFL thickness (0.81 ± 0.04). The parameter with the largest AUC obtained with the CSLO was rim area (0.72 ± 0.05), followed by rim volume (0.71 ± 0.05) and linear cup-to-disk ratio (0.66 ± 0.05). Temporal superior RNFL average thickness measured by SD-OCT performed significantly better than rim area measurements from CSLO (0.88 vs. 0.72; p=0.008). Outside normal limits results for SD-OCT parameters were associated with strongly positive LRs."
According to the news reporters, the research concluded: "The RNFL assessment with SD-OCT performed well in detecting preperimetric glaucomatous damage in a cohort of glaucoma suspects and had a better performance than CSLO."
For more information on this research see: Diagnosing preperimetric glaucoma with spectral domain optical coherence tomography. Ophthalmology, 2012;119(11):2261-9. (Elsevier - www.elsevier.com; Ophthalmology - www.elsevier.com/wps/product/cws_home/620418)
Our news correspondents report that additional information may be obtained by contacting R. Lisboa, Hamilton Glaucoma Center, Dept. of Ophthalmology, University of California San Diego, La Jolla, California, United States. Additional authors for this research include M.T. Leite, L.M. Zangwill, A. Tafreshi, R.N. Weinreb and F.A Medeiros (see also Eye Diseases and Conditions).
Keywords for this news article include: La Jolla, Glaucoma, California, Engineering, United States, Imaging Technology, North and Central America, Eye Diseases and Conditions, Optical Coherence Tomography.
Our reports deliver fact-based news of research and discoveries from around the world. Copyright 2013, NewsRx LLC